HDSL-1104 FOUR Fan Overhead ESD Ionizer for Static Control. Massive 440 CFM Output neutralizes static before static neutralizes your builds! THE most balanced ions for your money - in stock & ready to ship!
Our Heavy Duty HDSL-1104 4 fan overhead esd ionizer provides a massive ionic output of up to 440 CFM! Its balanced (+/- 5 V) ionic output makes it the perfect selection for large static control esd workbenches, static free work areas, production lines, packaging and more. Unlike many ionizers that require contaminant free air and continual maintenance, the SL-1104's steady-state, self balancing circuitry continuously adjusts to compensate for emitter point wear, line voltage fluctuations, variations in air particulate and velocity keeping maintenance to a minimum and performance at a maximum.
The HDSL-1104 features ultimate ionic output (but no overhead lighting like our HDSL-003). It's quiet, with fully adjustable airflow, long life tungsten emitters (typical life span four years and easy to replace) and a casing manufactured of a light weight aluminum alloy with a static dissipative powder coating.
4 Fan Overhead ESD Ionizer, Ultimate Ionic output
Outstanding quality, ISO manufactured.
POWERFUL performance, VERY low maintenance.
Certificate of compliance with each unit.
Precision + /- 5 volt steady state offset voltage.
Built in emitter cleaning brushes, easy to keep clean!
Self adjusting circuitry, factory calibrated.
Easily controllable high output multi speed fans.
Mounting brackets included with each ionizer.
CE Certified, fully warranted and built to last a lifetime!
The HDSL-1104 meets or exceeds the recommended technical requirements of ANSI/ESD S 20.20- 2014 tested in accordance with STM3.1.
HDSL-1104 Overhead ESD Ionizer
Price: $767.58 each
Free Shipping (Continental USA Only and details apply)
Call for quantity discounts!
Decay Test Results
Installation: The HDSL-1104 is an overhead unit designed to provide ideal coverage for control of electrostatics. Note: Do not insert objects into air intake or outlet grille and do not operate the SL-1104 in inflammable or explosive environments. The HDSL-1104 ionizer should be affixed 24 to 54 inches above the work surface with center targeted directly over work area. This unit may also be positioned and operated vertically should conditions dictate. Avoid drafty locations such as that caused by air conditioning, heater outlets, etc.
Attachment to overhead supports: S hooks and chains may be used for mounting these ionizers to structurally sound ceilings or wire shelves. Chains and mounting hardware should have a minimum safe working load of at least 50 pounds. When mounting the unit to ceilings it is important to check state and local codes to insure compliance. These units are typically hung in the same fashion as an overhead light fixtures (independently attached to the overhead building superstructure). Attachment to suspended ceilings only are typically not compliant with local codes. Mounting to workbench overhead supports: These ionizers may be mounted to work bench overhead shelves and beams using the supplied side mounting bracket in conjunction with angle iron, bolt on L brackets etc. These ionizers may be used either horizontally or vertically.
Electrical Requirements: The HDSL-1104 requires 110VAC /60Hz (standard) or 220VAC / 50Hz (optional) power. These units must be grounded. Each unit is supplied with standard US 3 prong male grounded power plug (standard). Do not modify the plug or use an ungrounded 3 prong receptacle. If an extension cord is required use heavy gauge fully grounded cords only.
Operating Procedure: Activate the HDSL-1104 overhead ionizing blower by turning the fan speed knob clockwise. The ionization indicator will illuminate and indicate the presence of ionized air. Airflow speed may easily be adjusted from 25% to 100% output utilizing the fan speed knob. The time required to neutralize electrostatic potential is dependent on the distance of ionizer to work surface, the voltage collected on the insulator and the speed of fans. When used in electronic assembly the ionized air stream should cover as much of the working area as possible. The constant flow of ionized air prevents items from developing an electrostatic potential and neutralizes electrostatic charges present on objects introduced to the airflow.
Routine Maintenance: Clean emitters points as needed. Inspect the points on occasion using a flashlight and magnifying glass. The points should be sharp and free of carbon build up and residues. To clean the ion emitter points press and turn the emitter cleaner knob (located in the center of each air outlet) clockwise one full turn and release the knob. The emitter points should always be sharp and not rounded at the point. Should these tips become rounded simply replace the emitter ring. Please Note: The typical emitter life is greater than 5 years. Replacement emitters rings are easy to install and available from United SCP. Periodically test the ionic output as indicated by your internal ESD program or ANSI/ESD S 20.20-2014, TR53, Air Ionizer Compliance Verification (discharge time and offset voltage)
United SCP Tames the Static Beast BEFORE it Takes Another Byte ®
© All rights Reserved